AmpFlSTR® Identifiler® Direct PCR. Amplification Kit for use with: reaction kit (Part no. ). reaction kit (Part no. ). Publication Part. Development and validation of the AmpFℓSTR® Identifiler® Direct PCR Amplification Kit: a multiplex assay for the direct amplification of. The AmpFℓSTR® Identifiler® Direct PCR Amplification Kit has been developed specifically to help address the DNA extraction and purification bottleneck that.
|Published (Last):||4 March 2007|
|PDF File Size:||8.24 Mb|
|ePub File Size:||4.34 Mb|
|Price:||Free* [*Free Regsitration Required]|
Thawing is required only during first use of the kit. To minimize the effect of instrument-to-instrument variation, identiffiler the same thermal cycler to amplify all three plates. Do not prepare all three plates simultaneously. Calculate the volume of each component needed to prepare the reactions, using the table below.
AmpFlSTR Identifiler Direct PCR Amplification Kit
Seed cells in plate format. The fluorescent dyes attached to the primers are light-sensitive. Increase the amount of DNA in the sequencing reactions. Periodically, optimizations and revisions are made to the kit More information. After first use, reagents are stored at 2 to 8 C and, therefore, do not require subsequent thawing.
See Allelic ladder requirements on page 28 for information on ensuring accurate genotyping. GeneMapper Software Version 4. Information in this document is subject to change without More information.
If you are processing multiple sample type and substrate combinations for example, buccal samples on treated paper idenrifiler buccal samples on swabsperform separate sensitivity experiments for each sample type and substrate to be used for testing. See page More information.
MBS For the quantitative determination of dirct free testosterone f-testo concentrations in identiifiler, plasma. All trademarks are the property of Thermo Fisher Scientific and its subsidiaries unless otherwise specified. See Table 1 on page 10 for a list of alleles included in the allelic ladder.
New Features in Version 5. All technical literature is available on the Internet at. To learn practical applications of DNA purification. This package insert must be More information. The data collection software stores information about each sample in a sample file. Not intended for any animal or human therapeutic. H May Add heat protocol for buccal swab lysate preparation. Data is therefore collected.
Data is therefore collected More information. Using this kit, PCR. Please refer to the User Guide of your automated or semi-automated disc punch instrument for proper guidance.
Centrifuge the plate at rpm for about identifileer seconds in a tabletop centrifuge with plate holders. G March Change copyright page information.
AmpFlSTR Identifiler Direct PCR Amplification Kit – PDF
Not intended for any animal or human. Find this course live.
There are two forms. This experiment accounts for instrument-to-instrument and sample-to-sample variations. Materials to be Supplied.